|Search By Accuracy|
Finding the right electrode for a specific application might at times look like a mind boggling exercise! In order to solve this problem, Hanna Instruments has divided the different electrodes into 7 main categories. You will then find the benefits of the two most common types of electrodes in each category together with a suggested part number. By following these charts, one can not only find the right electrode but also understand the reasons why one type is better than another.
Smaller (fits better on specially smaller or hand held meters).
Inexpensive (BNC connectors are widely used on TV sets).
Fastens better (due to the twisting mechanism).
Slightly faster (due to larger surface area of the bulb).
Easier to clean (larger bulb surface area).
Available in more shapes for special applications.
Stands higher temperatures (typically up to 100 Centigrade; plastic only up to 80 C).
More rugged (the electrode stem and bulb are well protected).
Costs less than glass electrodes.
Easier to clean and maintain. Sample trickles down to the tip. Best for applications such as dairy.
The tip is more rugged, hence better for penetration.
Can last longer.
A few seconds faster due to the larger surface area of the bulb.
Normally costs less.
Requires less maintenance. No refilling required.
Resists higher pressure (+15-30 psi).
Electrode could last longer if electrolyte is changed.
A different electrolyte, for example, silver-free or calomel for special applications can be used.
Slightly faster response due to greater flow of electrolyte into the sample.
Protects the sample from having any silver contamination from the electrolyte.
Almost no silver is deposited on outer ceramic as electrolyte is lost into sample hence no clogging from inside.
Protects the Silver wire from being contaminated due to prolonged use, or in highly acidic or alkaline samples.
Normally costs less. (The Hanna Double Junction Plastic Electrodes are however supplied at the same price as standard electrodes).
Allow a higher outflow of electrolyte, hence less likely to clog.
Recommended for low conductivity samples due to higher outflow of KCl.
Slightly faster response.
Stands higher temperatures (Typically 80°C or higher whereas open junction max. Temp. Is around 50°C)
Stands higher pressure.
No clogging (with a conic membrane becomes the ideal electrode for food samples with protein or glucose content or any medium which clog normal electrodes)